[time-nuts] 4 channel 1 Mhz - 25 Mhz distribution amp
kyrrin at bluefeathertech.com
Sun Jun 7 12:14:48 EDT 2009
I would be at least curious about the beastie. Please do post what you have.
*********** REPLY SEPARATOR ***********
On 07-Jun-09 at 12:06 wje wrote:
>I finally gave up trying to find a 50 ohm distribution amp I liked and
>wanted to pay for, so I built a simple 4 channel one that uses either a
>Linear LT1365 or an Analog Devices AD8044. It has an ac-coupled input
>and transformer-isolated outputs. I took care in the layout to keep
>trace lengths exactly the same for all channels, and they're also
>striplines. (not that using striplines makes much difference with 0.5"
>Warning - it doesn't have perfect unity gain, 0 chan-chan phase error,
>infinitely low noise, infiinte isolation, or any of those other
>characteristics Time-Nuts expect.
>What it does to (using the LT1365 @ 10Mhz) is accept a 10dBm input,
>provide 50dB worst-case channel-channel isolation, 68 dB worst-case
>output-input reverse isolation, and worst-case 0.2 degree
>channel-channel phase error. I'm not quite sure about the noise; it's
>down around the noise level of my HP 3588A, which reports -139dBm/rtHz
>for the amp. This is also the basic noise figure of the 3588, so the amp
>is probably lower. The amp specs say 9nV/rtHz, which works out to
>-147dBm referred to 50 ohms, assuming I did my quick calculations
>Interestingly, the phase error between 3 of the 4 channels is down
>around 0.02 degrees; just one channel has the 0.2 degree error relative
>to all the others. This was measured with my HP 5370 using 10k samples
>per reading. The LT1365 datasheet says 0.04 max. Hmm.. maybe I should
>check the coax for that channel.
>The gain is less than unity at all frequencies. Why? Because I wanted to
>be able to accept a 10dBm input without clipping and still get close to
>that out. Using the LT1365, this is possible. The AD8044 will start
>clipping at about 6dBm because it doesn't have enough current drive.
>However, it has a MUCH flatter bandwidth curve, managing 0.2 dB from
><1Mhz to >50Mhz. The LT1365 is significantly worse, but for a
>single-frequency distribution amp, this isn't all that important. Using
>the LT1365, the gain is -1.5 dB @ 10 Mhz, -0.9 @ 5Mhz. Of course, you
>can trim the gain for whatever you want within the voltage and current
>limits of the op amp. The opamp is set for a gain of 2 by default (to
>drive the series-terminated output transformer).
>If anyone is interested, I can put the schematic, pc board layout, and
>various plots from my 3588 for the amp on my FTP site.
>They said 'Windows or better'
>so I used Linux.
>time-nuts mailing list -- time-nuts at febo.com
>To unsubscribe, go to
>and follow the instructions there.
>__________ Information from ESET Smart Security, version of virus
>signature database 4136 (20090606) __________
>The message was checked by ESET Smart Security.
Bruce Lane, Owner & Head Hardware Heavy,
Blue Feather Technologies -- http://www.bluefeathertech.com
kyrrin (at) bluefeathertech do/t c=o=m
"Quid Malmborg in Plano..."
More information about the time-nuts