[time-nuts] OT favorite signal generator?
jmiles at pop.net
Fri Jun 19 09:57:19 EDT 2009
No distortion at low levels will occur with any signal generator worth
owning. Nothing from HP will do that.
-- john, KE5FX
> -----Original Message-----
> From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com]On
> Behalf Of optomatic at rogers.com
> Sent: Friday, June 19, 2009 6:39 AM
> To: Discussion of precise time and frequency measurement
> Subject: Re: [time-nuts] OT favorite signal generator?
> Hey Robert
> Great tip about the attenuator.
> I looked up some models on the internet and some look fairly expensive.
> I know that I will always be injecting low voltage signals, do you think
> it would be wise to buy a cheaper fixed attenuator, let's say 20dB?, and
> then just depend on the variable rate that the signal generator?
> Robert Darlington wrote:
> > I just bought an HP 3325A synthesizer/function generator that I
> really like
> > (for some things anyway) in about that price range. What you get is
> > probably dependent on what YOU need though. This thing is
> pretty limited
> > but this particular one has the high voltage option so the
> output goes up to
> > 40 volts up to 1MHz. It only goes to 20.99999999 Mhz (at up to 10V I
> > think) but for 99% of what I do that's just fine. It has a
> 10MHz external
> > reference which I hook to either an Rb osc or a Thunderbolt depending on
> > what I'm doing. It's pretty neat to see all the digits match
> between the
> > 3325A and the 5335A counter. Of course, that's being clocked
> by the same
> > oscillator too.
> > Also, I noticed on some signal generators, dropping the output
> down to a few
> > mV distorts the signal. What I learned to do was send my signal out at
> > about 1 volt ((RMS or p2p, doesn't matter) and go through a
> step attenuator
> > to drop it back down so it comes out clean.
> > -Bob, N3XKB
> > On Fri, Jun 19, 2009 at 6:04 AM, Patrick <optomatic at rogers.com> wrote:
> >> Hey everyone
> >> Sorry for the off topic post. I have received great advice in the past
> >> with items for my little shop and I can't resist to ask again.
> >> I am thinking about buying a signal generator. I suspect that I will
> >> mostly use it to inject low uV/mV signals into the amplification stages
> >> of the laboratory instruments I service.
> >> Any feedback you have would be greatly appreciated-Patrick
> >> _______________________________________________
> >> time-nuts mailing list -- time-nuts at febo.com
> >> To unsubscribe, go to
> >> https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> >> and follow the instructions there.
More information about the time-nuts