[time-nuts] Basic question regarding comparing two frequencies
lists at rtty.us
Mon Jul 26 19:43:16 EDT 2010
I've used the 6790 for this sort of thing before. It's a good choice since the whole signal chain is synthesized (if I remember correctly ..). It's still going to be tough to hit the originally requested accuracy with one.
On Jul 26, 2010, at 6:34 PM, Murray Greenman wrote:
> You guys are trying to crack a nut with a sledgehammer!
> For a start, as Didier says, you can't possibly read the frequency of a
> sky-wave signal to 0.01Hz in any short time frame since the Doppler on
> the signal can be as much as 1ppm (i.e. 10Hz at 10MHz). You can only
> infer it closer than that by studying the frequency in the very long
> In addition, you'll never know how much of the daily variation is
> ionospheric, and how much is due to thermal changes at the source.
> For what it's worth, the method I use for HF frequency measurements is
> much simpler. I use a receiver which I can lock to my GPSDO (RACAL
> RA6790/GM and HP Z3801A), and thereafter calibration is simply an issue
> of getting the sound card sampling rate correct at the software spectrum
> analyser, which you can do with a 1kHz reference from the GPSDO. No
> complicated signal generators, signal injection, or AM mode with AGC
> I use Peter G3PLX's SBSpectrum as the analyser, where you can trim the
> sample rate in tiny steps. It also has a frequency resolution of 25mHz,
> which is more than adequate for HF. My combination has won FMCs, but I
> still can't resolve 0.01Hz off-air.
> Whatever you do (with a sky wave signal) must be done over a long time
> frame in order to be sure of getting closer than 1ppm.
> Murray ZL1BPU
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts