[time-nuts] Fury - Rubidium
smace at intt.net
Thu Jul 29 00:29:08 EDT 2010
Recovery from holdover was fine as well.
On 07/28/2010 11:21 PM, SAIDJACK at aol.com wrote:
> That's an average accuracy of 6.6E-013 over 19 hours.
> In a message dated 7/28/2010 18:21:12 Pacific Daylight Time, smace at intt.net
> After about 19hours of holdover, only ~45ns. Now let's see how well it
> On 07/28/2010 01:02 AM, SAIDJACK at aol.com wrote:
>> Hi Scott,
>> a cool thing to try is to put the unit into manual holdover by issuing
>> The unit will act as if the GPS antenna has been removed, but GPSCon
>> continue to show the 1PPS phase drift against GPS. Thus you can see how
>> stable your LPRO is over time when the unit is in holdover.
>> You can restart normal locking with the command
>> On a good DOCXO, we would see less than 2000ns drift over a day when the
>> unit has been stable for a week or so. I believe the limits of the time
>> interval display are about +/- 2000ns.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
-------------- next part --------------
A non-text attachment was scrubbed...
Size: 21732 bytes
Desc: not available
More information about the time-nuts