[time-nuts] Heated crystal? & Rb tube corrosion (FE-5680A)
iteration69 at gmail.com
Thu Dec 1 01:00:44 EST 2011
I was starting to think i may have to crack open an instrument to get a
picture. But i found a reference online.
Note the corrosion around the cheaper metal parts (screws, spacers, shell).
I can't speak for the FE-5680A, but when i see something like this in the
instruments i maintain it's a tale-tale sign gas mitigation.
On Thu, Dec 1, 2011 at 12:16 AM, WB6BNQ <wb6bnq at cox.net> wrote:
> HUH ?
> What, exactly, do you mean by "corroded Rb tube interfaces ?"
> "Steve ." wrote:
> > snip
> > Also It appears that quite a few of
> > these have corroded Rb tube interfaces.
> > Thanks,
> > Steve
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts