[time-nuts] HP 10638A Degauser(s) available
paulswedb at gmail.com
Wed Nov 16 22:16:33 EST 2011
Bob good read and really appreciated the link.
I had a bad 004 tube so swapped in a standard tube. That tube wasn't in
great shape either but I learned a lot doing it.
So now I finally see what the magical degausser was all about. Clever how
they did it and could be reasonable implemented in newer technology. Though
I do like how they used relays for phase flipping at 2.4 seconds.
On Wed, Nov 16, 2011 at 9:53 PM, Robert Darlington <rdarlington at gmail.com>wrote:
> The manual says "Model 10638A Degausser is used with any HP 5061A Cesium
> Beam Standard which has Option 004 High Performance Tube installed.
> Degaussing is needed to obtain optimum performance from the Option 004 Beam
> On Wed, Nov 16, 2011 at 7:11 PM, jim s <jws at jwsss.com> wrote:
> > I have two of these beasties, which I'm lead to believe work with a 5061A
> > cesium standard.
> > I also managed to find and download the service manual / user guide for
> > I will offer them here first before ebay. Maybe they are useless, don't
> > know. Make a fair offer.
> > They were paired with HP laser rangefinder gear for some reason...
> > I don't think the guy had a 5061A which might be your next question.
> > Jim
> > ______________________________**_________________
> > time-nuts mailing list -- time-nuts at febo.com
> > To unsubscribe, go to https://www.febo.com/cgi-bin/**
> > mailman/listinfo/time-nuts<
> > and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts