[time-nuts] Fun with the Z3815
jmfranke at cox.net
Sat Aug 11 07:51:48 EDT 2012
Have you tried commanding a survey to get the receive to ignore the old
From: "Morris Odell" <vilgotch at bigpond.net.au>
Sent: Saturday, August 11, 2012 7:27 AM
To: <time-nuts at febo.com>
Subject: [time-nuts] Fun with the Z3815
> HI all,
> I posted a question recently here for a friend re a Z3815 not receiving,
> having formerly been perfect. Here's what he says now:
> " I've done more testing with my HP3815A. Firstly I rewired the coax cable
> between the external antenna connector and the PCB - there was some
> of snagging on the outside of the cable where it passes through the metal
> casing. However, no improvement in performance resulted. Also the blocking
> capacitor referred to turns out to be an inductor. DC is making it through
> to the antenna amp, and there is continuity along the whole route from
> N-connector to the Furuno GPS receiver input.
> The status screen on Hyperterminal shows that something is happening,
> however. The SYNCHRONIZATION section on the display showed outputs
> TFOM = 9 and FFOM = 3 (i.e. no sync). But the ACQUISITION section showed
> that although no satellites had been tracked, several had been detected -
> fact 9 of them. Also, the correct date and time were displayed. The LAT
> LONG data was incorrect, being the original northern hemisphere figures.
> Health monitor reads OK and there are no ALARMS. So the receiver appears
> be picking up enough information to show the satellites' PRN, EL and AZ
> furthermore they change between readings) as well as the date and time.
> What's more the time displayed is local time!"
> Any suggestions as to what's going on?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts