[time-nuts] T-Bolt Temperature
mblazer at satx.rr.com
Thu Aug 16 21:38:42 EDT 2012
Do you know the part number of a chip to replace the DS1620?
On 8/16/2012 5:58 AM, Arthur Dent wrote:
>> I have two Thunderbolts that I'm monitoring with
>> Lady Heather. The temperature on the older unit
>> (MFG 2/26/2004) seems to track the
>> environment. The newer one (MFG 11-24-2004)
>> shows 44.750000 °C and only changes in increments of exactly 1°.
> All REV-E Thunderbolts with date codes after about mid-2004
> used the DS1620 "E" chip and have this problem. Buyers have
> been told and believe the REV-E units are somehow superior to
> previous revisions so they will ask a known problem unit. I think
> they have seen articles on the internet that show really old units
> with the noisy Piezo OXCO and somehow believe that only the
> REV-E Thunderbolt has the newer Trimble OXCO, which is not
> true. After about 2001 all REV-A through E Thunderbolts had the
> new Trimble OXCO. I doubt you could find any with the older
> Piezo OXCO now.
> As far as I know there is only one U.S. seller of Thunderbolts on
> Ebay that replaces the DS1620 in these affected REV-E units
> with one that will display correctly with Lady Heather. I've heard
> several stories from buyers who have bought problem Thunderbolts
> from resellers, many of whom don't do any real testing to see their
> units work properly or are stable. Ask questions before you buy.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts