[time-nuts] Odetics 365
robkimberley at btinternet.com
Fri Oct 19 11:42:48 EDT 2012
I've just re-read this post after looking at the replies.
If the 365 is Time locked and showing and offset of 2.98E-13 then it looks
to working fine, and I would suspect that it is the frequency counter
oscillator that is drifting.
Is the frequency counter free running or locked to any other standard?
From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com] On
Behalf Of Rob S.
Sent: 18 October 2012 07:40
To: time-nuts at febo.com
Subject: [time-nuts] Odetics 365
Hello Group, first time post
I have just got my Odetics 365 working again after it being packed away for
a few years. I have it connected to a HP 5386A frequency counter and am
monitoring the 10 MHz output ( down to 0.01 Hz res ).
I am noticing the frequency is drifting about 1.5 to 2 Hz over the day and
settling low for a few hours ( aprox midday ) and then settling high about
12 hours later. The 365 is showing TFOM4 and Time locked, has a GDOP of 3.68
and displays a frequency error of +2.98E-013 and a Std error of Est of
4.80E-008. The averaging is turned off and it "see's" 3 to 5 satellites
continuously. My question is would this be normal behavior ( + or - 2 Hz )
for this type of equipment or is it looking like I have something that is
drifting a bit ? Also would switching the averaging on help ?
Any thoughts would be appreciated Group,
Thanks and Cheers,
time-nuts mailing list -- time-nuts at febo.com To unsubscribe, go to
and follow the instructions there.
More information about the time-nuts