[time-nuts] Once again about counter calibration
kb8tq at n1k.org
Sat Apr 25 08:41:27 EDT 2015
You would need to add another power supply into the 5386 case and rig it to always be on. You would also need to fit the MV89 in there. The
MV-89 pulls about 5X more power than the OCXO HP used in similar counters. It’s also roughly 5X the volume.
The simple / cheap / high reliability / lazy approach is to use an external reference.
> On Apr 24, 2015, at 11:46 PM, d0ct0r <time at patoka.org> wrote:
> Following my request, I am curious about "modding" of HP 5386A. My unit has TCXO. Is it possible to replace that existed TCXO by third part OCXO ? Like Morion MV89A, for example ?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts