[time-nuts] SMD TADD-1 distribution amplifier
bruce.griffiths at xtra.co.nz
Mon Dec 21 18:43:21 EST 2015
Such blanket statements aren't a particularly useful guide unless calibrated by measurements. For example the OPA653 has a measured PN floor of around -163dBc/Hz for a +13dBm input and the measured PN @1Hz offset is -150dBc/Hz (comparable with the NIST isolation amps).However the voltage noise is (estimated) to be 300nV/rtHz @1Hz and about 8nVrtHz @ 10kHz. Whilst the measured PN floor agrees closely with the measured value. The input voltage noise @1Hz can't be used directly to estimate the PN noise at 1Hz offset.
On Tuesday, 22 December 2015 12:02 PM, Charles Steinmetz <csteinmetz at yandex.com> wrote:
>How is that calculated? I only get this far:
>9.6nV/sqrt(Hz) into a 50R load is 1.8e-18 W/Hz or -147.3 dBm/Hz
As I said on Dec. 18 in response to the original post, the in-band
(10MHz) noise is NOT the main problem with respect to AM and PM
noise. The main problem is the BASEBAND noise (near DC, say,
0.1Hz--10Hz). And the AD8055, as well as the MAX477 originally used
in the TADD-1, are absolutely horrible in this regard. Both of them
have noise densities > 1,000nV/sqrtHz at 10Hz (and even worse below 10Hz).
See my earlier post for a brief explanation of why this is so, and
how baseband noise is converted to AM and PM noise in the RF signal
band. There are also more in-depth explanations in the
archives. One search term you can use is "PM conversion."
For low AM and PM noise, you want an opamp with a noise density of
10nV/sqrtHz, or lower, *at 10 Hz*. The ADA4899, LME49713, and AD8010
are three possible choices (these days, there are many others, as well).
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.
More information about the time-nuts