[time-nuts] PRS10 Rubidium
David C. Partridge
david.partridge at perdrix.co.uk
Sun Feb 22 15:45:39 EST 2015
Hmmm is it possible that a nearby board had a wet tantalum capacitor (H2SO4 electrolyte) that failed spraying acid?
From: time-nuts [mailto:time-nuts-bounces at febo.com] On Behalf Of Jürg Kögel
Sent: 22 February 2015 15:44
To: Discussion of precise time and frequency measurement
Subject: [time-nuts] PRS10 Rubidium
>I do not understand the corrosion since it has been in a very controlled environment in my basement.
More information about the time-nuts