[time-nuts] Help with my ADEV measurement setup
watsondaniel3 at gmail.com
Tue Jun 30 21:40:16 EDT 2015
Attached is a screenshot of the setup window. I manually typed in 1E-6 as
the units. I also hit Monitor and let it average the reporting interval for
a while until it settled at 1.00 seconds.
Originally I was leaving the time unit as 1, and microseconds in in the
serial string was not being detected to set the units automatically.
I'd be happy to send you a tim file as well if necessary. Let me know.
On Tue, Jun 30, 2015 at 8:51 PM, John Miles <john at miles.io> wrote:
> Let's see a snapshot of your acquisition dialog, just before you hit
> 'Start'. (Or send me a .tim file directly.) It can be tricky to configure
> the test setup for a TI measurement, since so many more things can go wrong
> compared to a frequency-based setup.
> -- john, KE5FX
> Miles Design LLC
> > -----Original Message-----
> > From: time-nuts [mailto:time-nuts-bounces at febo.com] On Behalf Of Dan
> > Watson
> > Sent: Tuesday, June 30, 2015 11:48 AM
> > To: time-nuts at febo.com
> > Subject: [time-nuts] Help with my ADEV measurement setup
> > Hi,
> > I'm trying to take some ADEV measurements with my 53131A, and I'm having
> > some issues. This is my setup...
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
-------------- next part --------------
A non-text attachment was scrubbed...
Size: 70040 bytes
Desc: not available
More information about the time-nuts