[time-nuts] Tait reference
GandalfG8 at aol.com
GandalfG8 at aol.com
Mon Jan 11 10:41:22 EST 2016
Are these the references with a rubidium oscillator ? They seem to have
similar models with OCXOs etc.
The T801 first seems to have hit the UK surplus market around 2005/2006, I
was told a batch of a couple of hundred, and units from that release were
still trickling through until around 2011 at least.
These were released by West Yorkshire Police when updating their UHF
mobile network and did contain rubidium standards.
The T801 contains dual nominal 12.8MHz synthesisers with very fine
switched preset adjustment for base station synchronisation. These are locked to
an external reference and can accept selected integer frequencies from 1 to
10Mhz and also 12.8MHz.
If the internal 10MHz rubidium module is fitted and used as the reference
then a short BNC link is required to couple the internal reference output
to the external reference input.
These were originally sold on Ebay with particular reference to the
rubidium module, for most purchasers the 12.8MHz synthesiser board would probably
have been of little interest, so I'm surprised to see no reference to the
rubidium module with the several recent auctions. This may be due to lack
of familiarity on the part of the seller but I certainly wouldn't want to
take anything for granted without asking.
I'm not sure either if the recent Ebay offerings are tail end leftovers
from that first batch or represent a more recent release, but I have seen a
number of failures with the rubidium modules in those bought between 2005
One at least was physically faulty with others refusing to lock, either at
time of purchase or subsequently, I have some of the latter still on the
ever growing "to do" pile.
According to the manual the rubidium module, where fitted, was originally
a Quartzlock TF4010A, but all units I've seen have been fitted with FEI
FE5660A modules. This required a degree of modification to the chassis and to
the heatsink and the quality of that work is often poor, so quite likely a
The unit also contains a DC-DC convertor to derive the 24 Volt supply for
the rubidium module from the 13.8 Volt input.
The lock on the front panel is just an ON/OFF keyswitch, if the key is
missing all those I've seen have been left in the ON position but it's easily
I've tried uploading my scan of the manual to Didier's site but gave up
after repeated page reset errors, it is now available here......
More information about the time-nuts