[time-nuts] Tbolt issues
kb8tq at n1k.org
Thu Sep 1 16:44:52 EDT 2016
I think one issue here is that ADEV is being used by one “lab" and absolute frequency is being used by the other.
They very much are *not* the same thing. There isn’t even a really simple way to convert one to the other.
There will always be a big delta between those two measures. For absolute frequency you will also need
some sort of confidence number (99 % or some such thing). The details of why all this happens are the
mainstay of the 1960’s (and early 70’s) papers on ADEV.
> On Sep 1, 2016, at 3:11 PM, Tom Van Baak <tvb at LeapSecond.com> wrote:
> Hi Bert,
>> because the frequency is constantly changed to correct time.
> A simple answer: you may have a bad TBolt. Was it part of the TAPR group buy, or did you buy it from eBay/China? If TAPR, you get a free replacement. Contact me off-list.
>> Tbolt is an excellent time device but not good for frequency reference past 1E-10.
> No, again it sounds like you have a bad TBolt. Or something is wrong (antenna? reception? time constant? environment? China resoldered parts?). I appreciate that Juerg did lots of testing -- do you happen to have his ADEV plot?
> I'm willing to help you debug this.
> (1) Attached is the ADEV of 8 random TBolts that I tested recently. How does this compare to yours? You can see mine are all under 2e-12 at 1 s and under 4e-12 at 100 s. On a bad day during holdover it might climb to 1e-11 at 1000 s but when locked GPS disciplining takes over and keeps it down to 2 or 3e-12.
> (2) For locked vs. unlocked TBolt ADEV, see http://www.leapsecond.com/pages/gpsdo/
> (3) Also attached is a frequency plot showing the typical noise and wander, down at the 1e-11 level. How does this compare with yours?
> Your claim of 1e-10 is order(s) of magnitude worse than the TBolts that I see. Something is wrong.
> ----- Original Message -----
> From: "Bert Kehren via time-nuts" <time-nuts at febo.com>
> To: <time-nuts at febo.com>
> Sent: Thursday, September 01, 2016 9:54 AM
> Subject: [time-nuts] Tbolt issues
>> We have been following the Tbolt power discussions but what I am missing is
>> the main problem with Tbolts. All the power work will not improve the
>> frequency performance of the unit because the frequency is constantly changed
>> to correct time. Tbolt is an excellent time device but not good for
>> frequency reference past 1E-10. I noticed it when I bought it and compared it with
>> my Tracor 527E on the needle and ever since used an Austron 2110 with a
>> digital 100 sec. loop for clean up. My Swiss partner Juerg has relied on an
>> OSA F3 for Tbolt clean up but continuous bad results on our work resulted in
>> a detailed analysis using a HP53132A counter and M100, FTS44060, two
>> OSA8600's and one of the best FE405's. The rsult is that the OSA F3 does not
>> clean up the Tbolt and we see +-4E-11 changes and old data shows even some +-8
>> E-11 excursions. With the popularity of the Tbolt an analog or digital
>> clean up loop would make sense. We are working on both, the analog because I
>> saw similar behavior on the FE5680 and FE5650, we did a GPSDO but do not plan
>> on using those Rb's but focus on M100 and FRK.
>> The collective expertise of time nuts could make a significant contribution
>> For power in critical applications we use the excellent work from Bern Kaa
>> a friend for the last twenty years and well known because of his published
>> work in the European HAM community
>> Bert Kehren
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts