[time-nuts] Cs standard stability
bob at evoria.net
Sat Mar 4 00:26:26 EST 2017
I hooked up my PRS-45A yesterday, just to do some ADEV comparisons with my GPSDO. Unfortunately, the GPSDO seems to be better by almost two points at all tau as measured on my HP 5370A and plotted by Timelab. So, can I infer from that that the OCXO in the PRS has a problem, or could it be something as stupid as a bad cap? Or do I just not understand the specs and the PRS isn't up to my expectations? The PRS docs show the unit as 3E-11 at 1s tau, and I'm seeing about 4.25E-11 at 1s, so maybe it's the latter.
NOTE: The test setup for this is 1PPS from a well-aged GPSDO to the 5370's start input, and 10MHz from either a different GPSDO or the PRS to the stop input.
As best as I can tell looking through the holes in the case (still running the tests), the PRS has an MTI 250-0827 5MHz OCXO. I've got one of the Lucent KS units that I don't care much about, so maybe the MTI 260-0624 5MHz in that will work? Both are 5MHz, but I don't know whether both are 12V at this point. And, of course, I don't know whether making the swap to an MTI 260 OXCO will benefit the PRS.
Any comments or educated speculations are welcomed.
Bob - AE6RV
More information about the time-nuts