[time-nuts] Early 5065A
paulswedb at gmail.com
Mon Mar 13 15:49:44 EDT 2017
Those are very nice pictures and I always appreciate detail ones.
Those caps are ugly but easily replaced and not to much damage that I could
Now with Corby's insights your in even better shape to revive the unit.
Best of luck.
On Mon, Mar 13, 2017 at 2:30 PM, <cdelect at juno.com> wrote:
> That is a VERY early unit.
> Those early ones did not have the TED device.
> I have only seen one unit with an earlier SN!
> Ohm the lamp oven windings to the threaded post protruding from the end
> of the optical unit. (should be open)
> Also check across the oven winding (50 Ohm)
> If the windings are toast (literally!) I can give you my instructions for
> removing and rewinding the oven.
> I also have spare oscillators and some other assys.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/
> and follow the instructions there.
More information about the time-nuts