[time-nuts] HP5061B Ion Current
pete at petelancashire.com
Mon Mar 20 22:44:00 EDT 2017
Do a search of the list and a few other sites. there is a LOT of
information on what challenges one can have.
And a lot of the information is practical.
As for as pumping a tube down, it took me pretty much a day + a whole
weekend, but it did go down to what was
scribbled on the door, and now takes less than a minute. I only turned mine
on either once every 2-3 months or when
I wanted to use it. Before the current issue of what looks like a power
supply going nuts, I got lock in no more that
203 minutes. If I was not going to use it, I would let it run for 30-45
minutes and shut it down for the next 'pump down'
On Mon, Mar 20, 2017 at 4:44 PM, Donald E. Pauly <trojancowboy at gmail.com>
> This is my first post. I just bought a non-working HP5061B on eBay
> for $350. My old boss KB7APQ in Salt Lake City and I are working on
> it. It came in from the Philipines in October of 2015 and was
> diagnosed with a bad physics package in March 2016 by AllTest in New
> Jersey. We initially found an open R8 pot on the 5 mc oscillator
> A10A2 which was killing most of the output.
> Next we had ion current of about 25 uA which stayed up. The book
> seems to indicate that this was low enough to enable the cesium oven.
> The analysis of the theoretical ion current for cesium turn on was
> very difficult. We shorted the base emitter of Q6 on A15 board to
> override Cesium oven disable. The instrument then achieved lock with
> beam current of 20. Ion current rose to 35 after cesium oven warm up
> and lock.
> We have made great progress on the +3500V power supply analysis. We
> believe that many so called gassy tubes are perfectly functional up to
> 1 mA ion current. Mean free path is on the order of 142 mm in that
> case or the length of the beam. We will discuss our findings if there
> is interest.
> WB4BBP has been most helpful with our efforts.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/
> and follow the instructions there.
More information about the time-nuts