[time-nuts] Holdover: Z3801A vs KS-24361
kb8tq at n1k.org
Thu Apr 12 08:58:47 EDT 2018
> On Apr 12, 2018, at 1:30 AM, Hal Murray <hmurray at megapathdsl.net> wrote:
> On that Status page, A Z3801A says:
> Predict 2.5 us/initial 24 hrs
That sounds about right for a 3801. The prediction is something better than
a wild guess and not what one would call a solid number.
> A KS-24361 says:
> Predict 394.1 us/initial 24 hrs
That would suggest the unit is unlocked. It should be below 10 us. Just as with
the 3801, the estimate is a bit fuzzy. You should be able to see issues in the data
on the Z3809/Z3810.
A prediction of < 1 can pop up on occasion, on either unit. More typical estimates
are in the “couple of us” range. If you yank the antenna, it still does a respectable
~5 us / 24 hrs or so either way.
> Both have been running for weeks.
> Is anybody seeing similar?
> Also, the Satellite Status area has a header of SS on the Z3801A and C/N on
> the KS. Does anybody know the units and/or what that is actually measuring?
There are tables that translate the Motorola (and some of the other brands) status
numbers into rough dbc/ Hz. I would not count on the numbers on a 20+ year old
design lining up directly with the latest and greatest modern module. Apparently
the SNR calculations are a bit weird.
> These are my opinions. I hate spam.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts