[time-nuts] EFOS2 Maser problem isolated! :)
paulswedb at gmail.com
Fri Aug 24 13:45:12 EDT 2018
Corby thathas got to be the most expensive fault detector I have run into.
But great to hear. Heck 5 MHz is DC.
On Fri, Aug 24, 2018 at 1:40 PM, <cdelect at juno.com> wrote:
> Well after a lot of monitoring found the 5Mhz isolation amp feeding the
> multiplier to have an intermittent output.
> Bypassed it using one of the iso/amps that feeds the front panel jacks.
> Parts on order before I pull it.
> I did make an interesting use of two HP5065A I have cooking on the bench.
> Needed to monitor two 5Mhz signals simultaneously from the maser for very
> intermittent dropouts.
> Fed each 5Mhz into the DIV cable from the A10 assy. on each 5065A and
> manually started the 1Mhz dividers.
> Then if a 5Mhz drops out the 1Mhz dies and wont return until reset.
> After two days caught the problem with the output of the iso/amp
> faulting its 5065A and the input to the iso/amp not faulting its 5065A!
> PIX of temperature controlled RF head and the two 5065A attached.
> time-nuts mailing list -- time-nuts at lists.febo.com
> To unsubscribe, go to http://lists.febo.com/mailman/
> and follow the instructions there.
More information about the time-nuts