[time-nuts] HP cesium beam tube life
paulswedb at gmail.com
Tue Jun 19 22:04:12 EDT 2018
Well one time nut did. Think its John and a ham K5??? call.
He wrote the process up using an external supply.
Thats why I have a 3KV supply on the shelf.
On Tue, Jun 19, 2018 at 8:43 PM, Perry Sandeen via time-nuts <
time-nuts at lists.febo.com> wrote:
> Yo Dudes!
> A while back someone asked about the life of a HP Cs tube.
> Fromthe HP 5061B Operators Manual:
> “2.7 The shelf life of the cesium beam tube I twoyears at storage
> temperatures up to +35 C (+95F) if the ion pump is operated 2 or 3 times a
> year. If the ion pump is not operated to maintainthe vacuum within the
> cesium beam tube, the expected shelf life is reduced.”
> “2.8 Temperature during storage and shipmentshould be limited as follows:
> a. Maximum temperature: +75 C (167F). Long term storage: +35 C (95F).
> b. Minimum temperature: - 40 C (-40).
> Whenstoring the HP 5061B for 6 months or longer, set the MODE switch to CS
> OFF andapply continuous as line power to the instrument. This enables the
> ion pump to maintain thevacuum within the cesium beam tube with the cesium
> beam tube off.”
> Bold emphasis by HP
> Mysnarky comment about 2.7. Shelf lifeinfers a tube uninstalled in
> storage. OK,how about a detailed procedure with drawings, for taking the
> tube and temporarilyrunning the ion pump to extend the shelf life?.
> Apparently no oneread this comment and thought about its logical
> time-nuts mailing list -- time-nuts at lists.febo.com
> To unsubscribe, go to https://lists.febo.com/cgi-
> and follow the instructions there.
More information about the time-nuts