[time-nuts] LT1016 as a pulse shaper...
kb8tq at n1k.org
Sun Mar 4 20:19:24 EST 2018
The “best” way to measure phase noise will always be a “that depends” sort of thing. One
pretty darn good way to check noise on any amplifier is to use something like a TimePod.
You use a power splitter and a pretty good source. First you check your TImePod (or whatever)
for floor. You then stick the amp in one leg. You re-measure phase noise (or ADEV). Assuming
it comes up above the previous floor, you have your number for the device. If it does not come
up above the previous floor, you need a different test set.
> On Mar 4, 2018, at 7:59 PM, Hal Murray <hmurray at megapathdsl.net> wrote:
> Bruce Griffiths <bruce.griffiths at xtra.co.nz> said:
>> If I had a suitable PCB board for it I would do the measurement properly.
> What would a suitable board look like and/or what sort of gear do you need to
> measure PN?
> These are my opinions. I hate spam.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts