[time-nuts] ✘NEO-M8N vs. NEO-M8T
Stromeko at nexgo.de
Mon May 21 14:41:23 EDT 2018
Gary E. Miller writes:
> Which I always thought was pointless, that only works for a fixed
> antenna. Any GPS in a fixed position lab will have a good rooftop
> antenna with clear skyview.
Except when it doesn't and then the ability to survive on fewer
visible/good satellites without going into holdover is most welcome.
> Except that requires a post process step, so not useful for real time.
No, you can very much use it to inform a consumer of the PPS in realtime
about the sub-quantization phase shift and have the PLL take this error
refinement into account.
> I just looked at the 'U-blox 8 Receiver Description' and it makes no
> mention of sawtooh anything. Is that in a different doc?
No, it's in there, look for the UBX-TIM series of messages, specifically
TOS and TP. However, it talks about offsets of the time pulse, not a
But there's a more specific description for series 6 timing receivers,
most if not all of which will be applicable to your series 8 module as
> I'll also test Surevey-In mode to see how much that helps.
_Any_ error in the surveyed position will show up as timing error that
also depends on the GPS constellation. I think Lady Heather provides a
special plot to map these deviations to the GPS sky tracks.
+<[Q+ Matrix-12 WAVE#46+305 Neuron microQkb Andromeda XTk Blofeld]>+
SD adaptations for Waldorf Q V3.00R3 and Q+ V3.54R2:
More information about the time-nuts