[time-nuts] Question about frequency counter testing
Oleg Skydan
olegskydan at gmail.com
Sun May 27 11:52:12 EDT 2018
Hi!
From: "Magnus Danielson" <magnus at rubidium.dyndns.org>
>>> You build two sums C and D, one is the phase-samples and the other is
>>> phase-samples scaled with their index n in the block. From this you can
>>> then using the formulas I provided calculate the least-square phase and
>>> frequency, and using the least square frequency measures you can do
>>> PDEV. The up-front processing is thus cheap, and there is meathods to
>>> combine measurement blocks into longer measurement blocks, thus
>>> decimation, using relatively simple linear processing on the block sums
>>> C and D, with their respective lengths. The end result is that you can
>>> very cheaply decimate data in HW/FW and then extend the properties to
>>> arbitrary long observation intervals using cheap software processing and
>>> create unbiased least square measurements this way. Once the linear
>>> algebra of least square processing has vanished in a puff of logic, it
>>> is fairly simple processing with very little memory requirements at
>>> hand. For multi-tau, you can reach O(N log N) type of processing rather
>>> than O(N^2), which is pretty cool.
>>
>> I had some free time today to study the document you suggested and do
>> some experiments in matlab - it was very useful reading and experiments,
>> thanks!
>
> Thanks for the kind words!
>
>> It looks like the proposed method of decimation can be
>> efficiently realized on the current HW.
I had some free time yesterday and today, so I decided to test the new
algorithms on the real hardware (the HW is still an old "ugly construction"
one, but I hope I will have some time to make normal HW - I have already got
almost all components I need).
I had to modify the original decimation scheme you propose in the paper, so
it better fits my HW, also the calculation precision and speed should be
higher now. The nice side effect - I do not need to care about phase
unwrapping anymore. I can prepare a short description of the modifications
and post it here, if it is interesting.
It works like a charm!
The new algorithm (base on C and D sums calculation and decimation) uses
much less memory (less than 256KB for any gaiting time/sampling speed, the
old one (direct LR calculation) was very memory hungry - it used
4xSampling_Rate bytes/s - 20MB per second of the gate time for 5MSPS). Now I
can fit all data into the internal memory and have a single chip digital
part of the frequency counter, well, almost single chip ;) The timestamping
speed has increased and is limited now by the bus/bus matrix switch/DMA unit
at a bit more then 24MSPS with continuous real time data processing. It
looks like it is the limit for the used chip (I expected a bit higher
numbers). The calculation speed is also much higher now (approx 23ns per one
timestamp, so up to 43MSPS can be processed in realtime). I plan to stay at
20MSPS rate or 10MSPS with the double time resolution (1.25ns). It will
leave a plenty of CPU time for the UI/communication/GPS/statistics stuff.
I will probably throw out the power hungry and expensive SDRAM chip or use
much smaller one :).
I have some plans to experiment with doubling the one shoot resolution down
to 1.25ns. I see no much benefits from it, but it can be made with just a
piece of coax and a couple of resistors, so it is interesting to try :).
All the best!
Oleg UR3IQO
More information about the time-nuts
mailing list