[time-nuts] hp 10544A and 10811A ovenized oscillators
mfeher at eozinc.com
Sat Nov 3 20:39:08 EDT 2018
Wow, I am surprised they allowed two weeks of playing around on an SSBN. I would have thought that on something like that the Navy would just replace it immediately. I was only in one once and I was real impressed by the professionalism of the crew. 73 - Mike
Mike B. Feher, N4FS
89 Arnold Blvd.
Howell NJ 07731
From: time-nuts <time-nuts-bounces at lists.febo.com> On Behalf Of Glenn Little WB4UIV
Sent: Saturday, November 3, 2018 7:10 PM
To: time-nuts at lists.febo.com
Subject: Re: [time-nuts] hp 10544A and 10811A ovenized oscillators
I had to "fix" a HP Cesium standard oscillator while on an SSBN.
The problem was a failed op amp in the control circuit for the crystal oscillator.
Due to lack of parts, the "fix" was to run the standard open loop with nothing controlling the crystal oscillator.
Using an oscilloscope to compare the good standard with the open loop standard, I I was able to adjust the failed standard oscillator.
It took over two weeks of adjusting the failed standard oscillator to keep the Lissajous pattern stable.
The oven was off for a total of less than an hour during troubleshooting.
Prior to failure the standard operated correctly for many months.
I was surprised as to the amount of time for the oscillator to settle down.
More information about the time-nuts