[time-nuts] RACAL-DANA 1992 Option 04E problems
danrae at verizon.net
Tue Jul 27 11:36:41 EDT 2010
billh40 at aol.com wrote:
> I own one RACAL-DANA 1992 (Standard- TCXO). I purchased one with the Option 04E on eBAY.
> The 04E chassis is sealed and the internal OCXO overheats due to lack of cooling air. This causes the counter to stop taking measurements. I had to add ventilation holes on the top and right side of the chassis lid to obtain reliable operation.
> Both counters work up to 1,750 MHz.
> Bill Hanna
Bill I have had several of these through my hands over the years and the
outside of the crystal oven does get hot but not exceptionally so.
Certainly it should /not/ require any extra ventilation. I would
suspect that you have a faulty oven heater controller in there and that
the heater is running at full current all the time.
What frequency does the standard put out? The frequency and the current
drawn by the oven should both settle down after about 15 minutes from cold.
Also if you really meant 1,750 MHz on both inputs that's way better than
it should be, pretty amazing really :^)
More information about the time-nuts