[time-nuts] RF isolation amplifier question
mfeher at eozinc.com
Sun Oct 14 12:32:52 EDT 2018
>From the schematic it does not look like anything special but only a 50 ohm
in and out circuit with fairly high isolation due to the common base of the
second BFT66. You may be able to simply use a newer MMIC from one of the
many sources to replace the assembly if you know the gain. Get one with more
gain than needed and then use a resistive attenuator for additional
isolation. I did not see the variable Johanson on the schematic and there
are no tuned circuits that I can see, at least not LC wise. I would think
you could measure the Johanson by removing it from the board. Since I do not
know the exact purpose of the amp I do not really understand why it has to
be so critical. Regards - Mike
Mike B. Feher, N4FS
89 Arnold Blvd.
Howell NJ 07731
From: time-nuts <time-nuts-bounces at lists.febo.com> On Behalf Of
cdelect at juno.com
Sent: Sunday, October 14, 2018 11:55 AM
To: time-nuts at lists.febo.com
Subject: [time-nuts] RF isolation amplifier question
I removed this amp from the EFOS2 Maser as it was intermittent.
Of course now I can't get it to fail on the bench!
I have fabricated a new PC board to replace it and have the components on
Note the 12pfd cap on the collector .
Some notes I have indicate is select at test and that is it a glass
The as built unit has a johansen 50pf that looks to be an early version of
surface mount with leads soldered on.
How do I determine what value to install on the new amplifier?
More information about the time-nuts